Before you When calculating rates.g. Quick Start-Up Guide Setting up an infusion for the Ivac Rate Calculation ml to be infused in 2 hours 30 minutes. The IVAC® Volumetric Pump automatically regulates the infusion rate of intravenous solutions. Users are advised to read, to understand this manual and to be thoroughly familiar with the Pump prior to Tel: (27) (0) interferon α-2α, , , , , , , interferon α-2b, , , , , IVAC, ixabepilone, –,, ,, , ixazomib, .
|Published (Last):||14 October 2013|
|PDF File Size:||20.81 Mb|
|ePub File Size:||19.43 Mb|
|Price:||Free* [*Free Regsitration Required]|
IVAC® PCAM® Syringe Pump
O pen HyperTerminal on your PC. Conditions of Use The Pump should only be operated by a clinician competent in use of automated infusion pumps and post-placement management of intravenous catheters. Page 14 This fluid does not enter the fluid path and requires no action. Should the clinical situation require it, the majual light can be disabled. Note the recommended joint position see illustration below.
Sale Lots Listing
All other components can be safely disposed of as per local regulations. Motor Drive and Speed control The motor speed is controlled by adjusting both the mark-space ratio of the drive signal and its repetition rate applied to the DC motor. Press to print patient history.
Starting with an empty, extended 50ml syringe on the pump, squeeze the finger grips and move the plunger holder slowly to the left. Charge retention will degrade over time. Fit item F – alarm cover label LB to decrease alarm volume if the alarm volume is still too loud at it’s lowest position.
D Remove the two screws which secure the battery retaining plate. Inspect the AC power supply plug and cable for damage. Alarm level for BD Plastipak only is 3.
The signal from the potentiometer is fed into the one of mqnual ADC inputs of the microcontroller. See Chapter 3, ‘Routine Maintenance’ for further information.
Remove the six case retaining screws and washers located on the base of the pump. Class A Class A EN Voltage Fluctuations, Flicker Emissions The pump is mnaual for use in all establishments, other than domestic, and those directly connected to the public low-voltage power supply network that supplies buildings used for domestic purposes.
Mains power quality should be that of a typical commercial or hospital environment. Spare Parts A comprehensive list of spare parts for this pump is included within the Technical Service Manual.
In normal operation if the watchdog fails the power is prevented from reaching the motor. Detach the retaining plate from the battery. Remove the three screws securing the flexi circuit to the carriage block. Remove the latch cam, O-ring and the rod from the case.
Check handset and handset inlet cable connections. R emove the three pole clamp screws and washers. The keypad is polled from the Control PCB.
CAREFUSION IVAC DIRECTIONS FOR USE MANUAL Pdf Download.
The flow sensor will cause mahual pump to alarm if a significant deviation from the infusion rate occurs. Check magnet iva been removed in test mode. To fit the spacer C remove the torsion rod by undoing the screw M3 x 6mm pan head and washer M3 wavy securing it to the plate.
If you wish to discard electrical and electronic equipment, please contact your CareFusion affiliate office or distributor for further information. Refer to the relevant DFU for comprehensive information. The user is responsible for verifying the suitability of a product used, if it is not recommended by CareFusion. Loading doses are not available.
Drugs and protocols use volume based units.